DI 140501 : VNG NysStar II
DI 140501 : NysStar II VNG Binocular system with goggles (hardware only). Select necessary modules for requested tests
The videonystagmography performed by a camera and infra red light measures and records the eye movement, which is analyzed by the software. NYSSTAR II is the essential tool to study the various functions of balance, namely the oculomotricity (ocular saccades, pursuit, optokinetic nystagmus) and the study of the vestibular system (spontaneous and positional nystagmi, rotary tests, caloric tests. NYSSTAR II is a nystagmography system and is able to record simultaneously both eyes movements, as well horizontal, as vertical. The calibration carried out by camera is stable and should only be made once at the beginning of the examination. The recording by camera removes all artefacts due to the changes of polarity between the electrodes and the skin like with the interferences of electric and electrophysiological origin.
NYSSTAR II is accompanied by powerful data-processing software, which simplifies the task of the doctor. According to the tests carried out, the software analyzes and measures various parameters:
number of jolts or ocular movements
slow phase speed of nystagmus
latency, phase, distortion
symmetry or asymmetry of the responses
paresis or labyrintic hypoexitability, directional
preponderance of nystagmus
comparison of the answers opened eyes, closed eyes and
measure of ocular fixation indices
THE NYSTAR II SYSTEM
NYSSTAR II is based on two USB cameras. Connected to a computer, they permit to localize the center of each pupil in real time. NYSSTAR II is provided in its case with its accessories: mask with two cameras, cap and USB interface.
Number of channels: up to 4 (horizontal and/or vertical, on each eye)
Resolution: 640 x 480 pixels on each eye
Sampling rate: 50 Hz - 200 Hz (option)
Connection: USB 2.0
100-240V - 50/60Hz 0.5A
Requires a computer following Difra's specification available on request
NYSSTAR II meets EN 60601-1 Standard and Directive 93/42/EEC.